What this meter does: The CONTOUR™NEXT LINK 2.4 meter offers a modern design with multiple features for greater ease of use and discretion.
Who this meter suits: Medtronic MiniMed™ 640G users who want to send blood glucose results wirelessly to their insulin pump.
Insert the grey end of the CONTOUR™NEXT test strip into the meter. The meter will turn on.
When the Apply Blood screen appears, touch the test strip to the drop of blood. Hold it there until the meter beeps.
Your test result will appear after a 5 second countdown. To turn the meter off, press and hold the top menu button. (The meter also turns off after 3 minutes of inactivity).
Just follow the 6 easy steps below and start sending test results wirelessly to your pump.
From the Setup menu go to Pump Options and press OK. The Pump Options screen appears and asks if you would like to Connect now?, press Yes, then press OK.
Pick up the pump. From the Menu screen select Utilities, then select Device Options, then select Connect Device.
Place the meter and pump next to each other. Select Auto Connect†on the pump and on the meter. Be sure other nearby Medtronic devices are NOT in search mode. Scroll down and select Continue.
Keep the meter and pump next to each other. Select Search on the pump and on the meter. The search may take up to 2 minutes.
Check that the serial number on the pump matches the serial number on the meter. If they match, scroll and select Confirm on the pump.
The screens on both devices should confirm that meter and pump are connected.
CONTOUR™ meters will display blood glucose measurements in your country's standard metric (mg/dL or mmol/L).
*For full details regarding use, please refer to the system user guide provided with the meter.
*For additional helpful information, please refer to the system user guide provided with the meter.
†With Auto Connect, the meter automatically searches for a pump. With Manual Connect, you enter your meter serial number into the pump. See user guide for information about Manual Connect.